Introduction? SMA attenuators are used to reduce the input power, and therefore enable RF Explorer to measure stronger signals. It is also used to protect the input SMA port of RF Explorer when unknown strong signals may be in the environment, so you can use this attenuator on demand. You can easily unplug the attenuator when measurement is for weak signals. You can also daisy chain multiple attenuators to get additional values. For instance a 10 dB + 30dB attenuators connected in series will give you 40dB total attenuation.Â Â RF Explorer firmware includes an âOffset DBâ feature in the Attenuator Menu, you should specify (+010) dB in this setting when the attenuator is connected in order to get correct readings on screen.Â This powerful 10dB attenuator will work reliably up to 6GHz and 2W input power.Â See this article for more details on protection of your RF Explorer.Â Â Feature Reliable V.S.W.R Easy maintenance Wide broadband frequency applicationÂ Directional 2 Watt rating Â Specification Frequency range: DC - 6.0 GHz Impedance (Nominal): 50? Power rating: 2 Watt (+33dBm) Attenuation: 10 dBÂ Tolerance: +/- 0.5dB/max V.S.W.R (Max): 1.22 : 1 Connector type: SMA male â female Â For any technical support or suggestion, please kindly go to our forum.
If you are interested in Internet of Things (IoT) or M2M projects check our open
source sensor platform Waspmote
which counts with more than 100 sensors available to use 'off the shelf', a complete API with hundreds of ready to use codes and a low consumption mode of just 0.7µA to ensure years of battery life.
Know more at:
Get the Starter Kits at: